Meter info for anyone owning one:
We purchased two of these meters and found the same fault with both..
The condition is cont buzzer intermittant and ohms zero float..
Found the fault to be the lead post securing screws on the board [three small phillips] to be loose, causing poor contact..
Oddly enough, both were the center [ neg/com. ] post..
Do secure them, but do not over tighten...
If your cont buzzer is not steady and/or erratic, take a look at this connection...simple test is to connect leads together and go to cont. buzzer and jiggle the leads at the post ends.. you will hear it through the piezo....looking for steady tone if good connection.
Otherwise, fine product for the $$$$..
Maybe we got the Coffee Break run...
Last edited by Arthur Dalton; 06-30-2003 at 10:05 AM.