is there a way to open up the old probe without damaging the tube? i was going to slit the tube covering and i was trying to get the tip off but wasn't successful. i remembered that i have some thermal paste that i used on a CPU - heatsink contact so i'll use that. RS just gave me the usual blank looks and i didn't feel like checking out their entire inventory (yet again)...
ps. a 356? cool! how about some pix?
1993 300e-2.8 - gone now <sigh>
"Do not adjust your mind, it's reality that's malfunctioning"